The compact pressure moulding unit MINISTAR S® equipped with scanning technology: scanner reads the coded material, allowing programming of all important parameters such as heating time, temperature and cooling time. In other words: precise pressure moulding results are reached in record time thanks to the extremely fast, patented heater and working pressure of 4.0 bar, adjustable via pressure regulator. New generation equipped with user-friendly membrane keypad and large display. All operating parameters are monitored and indicated on the display.
No, you can only process material with a diameter of 125 mm in SCHEU-DENTAL pressure moulding machines. Make sure to use the correct material size to avoid any problems during pressure build-up. The rim of a too small blank gets pulled out from the clamping of balancing and locking ring. Thus, a precise and correct thermoforming result is not ensured.
Maybe there is a pellet on the edge of the model cup between material and clamping ring. Interrupt the thermoforming process, taking care the model cup rim is absolutely clean before closing the pressure chamber.
Leaking of air usually does not affect the thermoforming result, because the compressor is constantly supplying air. Nevertheless, you should check the following:
When working on the model platform,
When embedding the model into the pellets, sharp model edges may cause perforation mainly of thin material.
No, this is not possible. However, depending on the series of your pressure moulding device, you can set an individual heating time using the “clock” symbol on the keyboard of your device or by using the “+” and “-” buttons. Please refer to the manual of your device to find out which option applies to your device. The operation manual of the current series of our pressure moulding machines can be found on this site under "Further Information" for download. Operation manuals of previous device series are available under SERVICE on top of this web site.